Conjugate Mismatch-Loss Chart.
relationships between the impedances of the source and thermistor mount. This
uncertainty is +0.30 dB to -0.28 dB and can be determined from the Mismatch
Loss Limits charts in Figures 4 and 5.
(6) The result of the total mismatch loss uncertainty on the Zo
available power leave is determined by algebraically adding the thermistor
mount loss to the uncertainty caused by source and thermistor mount Zo mismatch
SWR. Thus, the Zo available power uncertainty is (-0.07 dB) + (+0.30 dB), and
(-0.07 dB) + (-0.28 dB), equal to a range of +0.23 dB to -0.35 dB to +5.5% to -
8.2%. The power delivered by the