Expanded Conjugate Mismatch-Loss Chart Covering
SWR's from 1.00 to 1.20. Chart is read in same manner as
source to a Zo load, with source and thermistor mount mismatch as in this
example, would be somewhere between 0.23 dB (5.5%) below the maximum power and
0.35 dB (8.2%) above the minimum power actually entering the thermistor mount.
(7) Power measurement uncertainty caused by mismatch loss is one source
of error to consider when measuring Zo available power without a tuner.