Flow Chart for the US Army Volt
disseminate, and assign the values to the AACL voltage standards.
(2) The AACL using saturated standard cells, will obtain and transmit
the data as required to implement the program.
The full technical procedure is contained in TB 750-245-
(1) Calibration of Cells at the AACL. Program schedules, methods of
shipping the transport standards, observation forms, and data transmission
instructions will be controlled by the USASL to assure calibration of the
AACL standard cell as required by TB 43-180.
Program intervals may be
shortened to satisfy emergency needs or to meet individual requirements.
(2) Upon receipt of the transport standard at the AACL, it shall be
inspected while on battery power to ascertain that it is maintaining
operating temperature (periodic cycling of front panel