(r) Repeat technique of (i) through (q) above for each TI model at
temperature-bath temperatures listed in table 8. If average allowable
deviations are not within limits specified, perform b below.
(a) If TI is out of tolerance according to average allowable deviation
column of table 8, but within tolerance according to rejection tolerance
column, prepare calibration curve, using values from paragraph 7 b (2)
and a above and attach to TI.
(b) If TI is out of tolerance according to both average allowable deviation
and rejection tolerance columns of table 8, TI must be rejected.
i. FINAL PROCEDURE
(1) Disconnect all equipment.
(2) Rinse and dry TI and standards and return them to their protective cases.